Nose/throat-swabs from 1049 patients were screened for MRSA using CHROMagar MRSA, LightCycler Advanced MRSA, and Detect-Ready MRSA. Results were compared to the CHROMagar MRSA results, which was set as reference system. MRSA was detected in 3.05% of the patients with CHROMagar MRSA. LightCycler MRSA Advanced showed a higher clinical sensitivity (84.38%) than Detect-Ready MRSA (57.69%).The negative predictive values were high for both tests (>98%). The specificity and the positive predictive value were higher for the Detect-Ready MRSA test than for the LightCycler MRSA test (99.59% and 78.95% versus 98.52% and 64.29%). For routine screening LightCycler MRSA Advanced proved to be more efficient in our clinical setting as the clinical sensitivity was much higher than the sensitivity of Detect-Ready MRSA. CHROMagar MRSA detected more MRSA positive samples than both PCR methods, leading to the conclusion that the combination of PCR with cultural screening is still the most reliable way for the detection of MRSA. LightCycler MRSA Advanced was faster and needed less hands-on time. The advantage of Detect-Ready MRSA was the additional identification of methicillin-sensitive S.aureus (here in 34.63% of the samples), an information which can be possibly used for reducing the risk of postoperative infections in surgical patients in future.
Citation: Aydiner A, Lüsebrink J, Schildgen V, Winterfeld I, Knüver O, et al. (2012) Comparison of Two Commercial PCR Methods for Methicillin-Resistant Staphylococcus aureus (MRSA) Screening in a Tertiary Care Hospital. PLoS ONE 7(9): e43935. doi:10.1371/journal.pone.0043935
Editor: Herminia de Lencastre, Rockefeller University, United States of America
Received: May 4, 2012; Accepted: July 27, 2012; Published: September 19, 2012
Copyright: © Aydiner et al. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Funding: The authors have no support or funding to report.
Competing interests: Oliver Schildgen reports to having received a travel grant by Roche Diagnostics, Germany. No other competing interests apply to this manuscript. This does not alter the authors' adherence to all the PLOS ONE policies on sharing data and materials.
Nosocomial infections caused by methicillin-resistant Staphylococcus aureus (MRSA) have been shown to be associated with severe clinical and economical outcomes – Whereas high MRSA prevalences were reported from countries like the USA, Taiwan, Japan, or the southern European countries (e.g. up to 80% of methicillin-resistant S. aureus of S. aureus in blood-cultures) the prevalences in the Netherlands or Scandinavia were low (<1–5%) , , , , . These low prevalences appear to be the result of a more effective MRSA management , , consisting in MRSA screening at patient admission, isolating colonized patients in single-rooms or cohorting them, decolonization of MRSA-positive patients, and strict staff hand hygiene , .
However, screening methodology is crucial for detecting MRSA and no method is tested the most sensitive, specific or cost effective, so far. Inexpensive culture methods have the disadvantage of a swab-to-result time of up to 48 hours and different commercially available media show even differing accuracy values .
More expensive molecular-based test systems show more interesting turn-over-times but only few studies estimating accuracy values in a routine diagnostic setting are available so far . Data of the Detect-Ready® MRSA Kit concerning these questions are still lacking.
In our study we compared the Detect-Ready® MRSA Kit (MDI, Kent, UK) with the standard methods used in our laboratory, the LightCycler® MRSA Advanced Test (Roche, Mannheim, Germany) and the CHROMagar MRSA II (BD, Heidelberg, Germany). Our screening comprised all patients at admission to our emergency department or two intensive care units in accordance with our routine MRSA screening management.
Materials and Methods
Settings and specimen collection
The study was conducted at the Kliniken der Stadt Köln, a 1500 bed tertiary care facility and university affiliated teaching hospital in the City of Cologne, Germany. During a four month period from August to November 2011 all patients admitted to two intensive-care units or the emergency department of the hospital were screened for MRSA and included into the study, leading to 1049 patients included. Specimens were collected by the nursing staff using double-headed swabs with amies gel (Copan, Italy). In each patient a combined swab was taken from the throat and both nares, the most important sites for MRSA colonization , , , . Technically, the throat was swabbed rotating against the mucosa using a double-headed swab which was then used for swabbing both nares in the same way. The aim of the study was to evaluate two diagnostic assays for potential routine use, thus no ethical vote was required. The LightCycler® Advanced MRSA Assay (Roche) was performed as the routine diagnostic assay in our laboratory, whilst the Detect-Ready® Assay (Molecular Detection Inc) was performed in addition during the study period. Therefore verbal informed consent was sufficient. All procedures were performed according to the declaration of Helsinki in its present form.
Laboratory processing and culture methods
The BBL CHROMagar MRSA II (BD, Heidelberg, Germany) was used as screening agar for MRSA . Additionally, Columbia-Agar with 5% sheep blood (BD) was used as control for bacterial growth and for identification of methicillin-sensitive S. aureus (MSSA) (Figure 1). In the laboratory both swabs of the double-headed swabs were firstly separated and each part was used for inoculating one MRSA selective agar and one Columbia agar consecutively. The two parts of the swab were then used for DNA-extraction for the one or the other PCR method, one for the LightCycler® MRSA Advanced Test, one for the Detect-Ready® MRSA Kit. The culture plates were incubated at 36°C and evaluated after 24 and 48 hours. Mauve-coloured colonies onto the selective chromogenic agar were confirmed to be S. aureus using an agglutination test for the simultaneous detection of the fibrinogen affinity antigen (clumping factor), protein A, and the capsular polysaccharides of S. aureus (Pastorex Staph-Plus, Bio-Rad, Munich, Germany). In case of agglutination the culture was sent to the routine diagnostics microbiology laboratory serving the Kliniken der Stadt Köln for species identification and antibiotic-resistance testing via the Vitek2 system (bioMerieux).
Figure 1. Clinical experimental design.
Nasal/throat specimens were tested by Detect-Ready® MRSA Kit and by the LightCycler® MRSA Advanced Test. For testing, the double-headed swab was separated. One swab head was processed for directly plated culture on BD BBL™ CHROMagar™ MRSA Medium II, BD Columbia Agar with 5% sheep blood plates, and the LightCycler® MRSA Advanced Test. The other swab was used for the assay with Detect-Ready® MRSA Kit and directly plated culture on BD BBL™ CHROMagar™ MRSA Medium II, and BD Columbia Agar with 5% sheep blood plates. MRSA positive colonies onto the selective chromogenic agar were confirmed to be S. aureus using an agglutination test.doi:10.1371/journal.pone.0043935.g001
In case of the absence of typical (mauve coloured) colonies on the chromogenic medium and colonies suspected to be S.aureus on the Columbia agar, these colonies were analysed in the same way.
The LightCycler® MRSA Advanced Test targets the integration site of the SCCmec cassette into the S. aureus chromosome (SCCmec:orfX junction). The Detect-Ready® MRSA Panel Kit simultaneously targets three MRSA DNA regions, namely the mecA gene, the nuc gene and SCCmec:orfX junction. In this way the assay is able to identify MRSA, MSSA and Coagulase negative staphylococci (CoNS) with the help of integrated software considering the different amplification rates. CoNS are assumed if the mecA gene is detected whereas the nuc gene and the SCCmec:orfX junction remain undetectable. Both, the LightCycler® MRSA Advanced Test and the Detect-Ready MRSA Panel Kit (Rotor-Gene compatible version, MDI, Kent, UK) were performed according to the manufacturer's instructions. DNA was extracted using the LightCycler Advanced Lysis Kit (Roche) for the LightCycler MRSA Advanced Test and the Detect-Ready MRSA Swab Lysis Kit (MDI) for the Detect-Ready MRSA Panel Kit according to the manufacturer's instructions.
Each PCR method contains an Internal Control (IC) to exclude or detect PCR inhibition and to monitor reagent integrity. The IC must be detected positive in all MRSA negative specimens otherwise the specimen is automatically set as “invalid” by the program software.
A true positive (TP) result was defined as positive result in the PCR and the corresponding culture. A true negative (TN) result was defined as a negative result in the PCR and the corresponding culture. A sample positive in the PCR and negative in the corresponding culture was defined as false positive (FP), a sample negative in the PCR and positive in the corresponding culture was defined as false negative (FN).
Cultured MRSA isolates from samples with negative PCR results were reanalyzed by PCR. Isolates with repeated negative result were genotyped with Identibac StaphyType (Clondiag, Jena, Germany). The analysis was conducted by Alere Technologies GmbH (Jena, Germany). In addition the SCCmec type was determined according to the method of Boye et al. .
The results of the LightCycler® MRSA Advanced test and the Detect-Ready® MRSA test were compared to the corresponding CHROMagar MRSA plates. The clinical sensitivity, specificity, and the negative and positive predictive values (NPV and PPV) were calculated for each PCR method. Samples with invalid PCR results were excluded from the calculations.
Patient profiles and culture results
Of the 1049 patients, 499 (47.57%) were female and 550 (52.43%) were male. Overall, 214 (20.4%) patients were positive for S. aureus (95 female, 119 male), of which 32 were MRSA isolates (14.95%). The MRSA rate in the whole cohort was 3.05%).
31 of the CHROMagar MRSA plates inoculated with the swabs used for the LightCycler® MRSA Advanced Test were positive for MRSA. Only 27 of the plates inoculated with the swabs used for the Detect-Ready® MRSA Kit were positive for MRSA. This difference is presumably due to improper sample collection technique leading to different amounts of material on the two heads of the swab. Only the samples tested positive for MRSA in the corresponding chromagar culture were included in the determination of true positive and false positive PCR results, except two cases with a positive MRSA result in both PCRs, but only on one of the culture plates. Those samples were defined as true positive results in the PCR as well, resulting in 32 culture positive samples. Of the 1049 samples, 21 samples could not be analyzed with the Detect-Ready® MRSA Kit, including two of the positive samples, due to a limited number of test kits.
The LightCycler® MRSA Advanced test detected MRSA in 42 samples, 27 of these samples were also positive on the CHROMagar, resulting in 27 true positive and 15 false positive results. Of the 1005 samples tested as MRSA negative in the LightCycler® MRSA Advanced test, 5 samples were positive on the CHROMagar and classified as false negative results (Table 1).
Table 1. Comparison of PCR results.doi:10.1371/journal.pone.0043935.t001
With the Detect-Ready® MRSA Kit only 1028 samples could be processed. The test identified 19 samples as MRSA positive, 15 of those samples were positive on CHROMagar and therefore classified as true positive, the other four samples were classified as false positive. The other 989 samples were tested as MRSA negative; in detail 173 were identified as MSSA, 183 as a mixture of MSSA and coagulase negative staphylococci (CoNS), 324 as CoNS, and 309 as negative (Table 2). Ten of the MSSA and CoNS positive and one of the CoNS positive samples showed MRSA growth on CHROMagar and were classified as false negative in the PCR (Table 1).
Table 2. Results of the Detect-Ready MRSA PCR.doi:10.1371/journal.pone.0043935.t002
Overall 22 specimens lead to invalid results upon testing with the LightCycler® MRSA Advanced Test (2/1049) and the Detect-Ready® MRSA Test (20/1028). All specimens were negative on CHROMagar MRSA and were not included in the calculation of clinical sensitivity, specificity, PPV, and NPV of the respective test.
The LightCycler® MRSA Advanced Test achieved a better clinical sensitivity than the Detect-Ready® MRSA Kit (84.38% versus 57.69%) in our clinical setting. The low sensitivity of the Detect-Ready® MRSA Kit was basically due to a number of samples falsely identified as a combination of MSSA and CoNS. The negative predictive values were nonetheless high (>98.5%) for both tests. The specificity and the positive predictive value were higher for the Detect-Ready® MRSA Kit (99.59% and 78.95% versus 98.52% and 64.29% in the LightCycler® MRSA Advanced Test) (Table 1).
Samples with MRSA growth on CHROMagar and negative PCR results were reanalyzed. The cultured MRSA isolates retrieved from the CHROMagar plates were used as templates for the respective PCR. In this approach three of five isolates from samples with false negative results in the LightCycler® Advanced MRSA-PCR were tested positive, one negative. One cultured sample was missing and could not be reanalyzed. The isolate tested negative was MRSA positive in the than performed Detect-Ready® MRSA Test. Concerning the Detect-Ready® MRSA PCR, the analysis of the isolates from 11 false negative PCR samples resulted in eight MRSA positive and three negative PCRs (one negative, one CoNS, one MSSA+CoNS). All three negative samples were tested positive in the LightCycler® MRSA Advanced Test. The four samples that failed to be detected in one of the PCR assays were sent to Alere Technologies GmbH for genotyping. All four MRSA strains belonged to the ST5/ST225-MRSA-II strain. SCCmec typing revealed that all had the SCCmec-cassette type II.
MRSA is still a growing problem in health care settings leading to increased costs and patient risks. Identification of colonized patients is the first step in the containment of MRSA spreading. Different diagnostic tests are available for the identification of colonized patients so far.
Factors to consider for the choice of a MRSA screening platform include sensitivities, specificities, turnaround time, costs, and ease of interpretation, which had been shown to vary considerably . MRSA screening methodology is already widely in use even though accuracy values of the respective tests are not all sufficiently tested in a clinical routine setting or even in comparison to already better evaluated tests.
Here, we compared the Detect-Ready® MRSA Kit (MDI) with the methods used routinely for MRSA screening in our hospital (LightCycler® MRSA Advanced Test (Roche) and CHROMagar MRSA (BD)).
Costs and turnaround time play an important role in the decision which assay is the test of choice. Reagent and instrument costs are much higher and turnaround times shorter for the PCR assays. LightCycler® MRSA Advanced Test had the shortest turnaround time with less than 2 hours. Processing of the Detect-Ready MRSA Test was finished within 5 hours requiring a considerable hand-on time. Results from cultures were available after 24 hours of incubation at the earliest (94.78% of MRSA positive cultures could be identified after 24 hours (Table 1)). Interpretation of positive results was easy to perform for both PCR platforms, with software programs providing clearly arranged result lists after every PCR run
Both test methods are based on the detection of the SCCmec-orfX junction. The staphylococcal cassette chromosome mec (SCCmec) is the carrier of the resistance-gene mecA. It integrates into the S. aureus genome in the region of the open reading frame orfX which is specific for S. aureus . Therefore, an amplification product is obtained in SCCmec carrying S. aureus but not in SCCmec carrying CoNS. SCCmec PCRs have been reported to have high sensitivities and specificities and are used in several commercial available MRSA detection kits , , , , .
In this study both tests had high specificities (LightCycler® MRSA Advanced Test: 98.52%, Detect-Ready® MRSA Kit: 99.59%). LightCycler® MRSA Advanced Test was more sensitive for the detection of MRSA (84.38%) than Detect-Ready® MRSA Kit, but both tests had poorer sensitivities in our real-life study setting than reported in previous studies . Especially the Detect-Ready® MRSA Kit had a surprisingly low clinical sensitivity (57.69%). This might be due to a software/cut off problem of the Detect-Ready MRSA test, as the software calculates automatically if MRSA or a mixed population of staphylococci is present in the samples (10 of the culture positive samples were identified as MSSA and CoNS and one as CoNS). After re-analysis of the false negative results using the cultured strains as PCR template, three of the five false negative LightCycler® Advanced MRSA Test results and eight of the 11 false negative Detect-Ready® MRSA results tested positive, which would increase the sensitivities so 93.75% and 88.46%, respectively, if the samples were tested positive in the first PCR.
While the Detect-Ready® MRSA PCR detected more false negative results, the LightCycler® MRSA PCR produced significantly more false positive results (LightCycler® MRSA: 15 FP, Detect-Ready® MRSA: 4 FP, p<0.05) resulting in a lower predictive positive value (64.29% versus 78.95%).
The detection of false positive and false negative results could have had several reasons. Recent studies reported a S. aureus strain (LGA251) resistant to methicillin but negative for mecA , . This strain harbours a divergent mecA homologue with a different organization than other SCC elements leading to a false negative result. Another study from Denmark  revealed that a specific common SCCmec clone was frequently not detected in a commercial MRSA assay leading to the conclusion that local diversities play an important role in the performance of MRSA assays, as undetectable low prevalence strains could become widespread among S. aureus. In our study, most of the samples which were false negative in the PCR proved to be positive in a second PCR approach using the isolated MRSA cultures. In those cases most probably an inoculum effect (plates were inoculated prior to MRSA PCRs), low concentration on the epithelium of MRSA, or inhibitory PCR effects seem to be the cause for retrieving false negative results in the PCR assays. Four samples were tested negative again in the second PCR approach, but all samples were MRSA positive in the conquering assay. Genotyping revealed that all strains belonged to the ST5/ST225-MRSA strain with a SCCmec-cassette type II, a common type in this region. Future sequence analysis of these four strains would be interesting to determine why they could not be detected by the PCR assays. Minor changes in the sequence of the primer binding sites could be the reason. Maybe there is a MRSA subpopulation in our region which is not detectable with certain commercial assays comparable to the MRSA clone in the above mentioned Danish study .
The reason for false positive PCR results can be orfX genes in CoNS homologue to the S. aureus variant or SCCmec cassettes lacking mecA , , , . To avoid false positive results in this case, the Detect-Ready® MRSA Test detects a second S. aureus specific marker (nuc) and the mecA gene additionally to the SCCmec-orfX amplicon. Overall, this approach enables the differentiation between MRSA and MSSA in addition. This may be important in surgical patients, as preoperative detection of MSSA nasal carriage and decolonization of the patient could reduces the risk of surgical site infections due to S. aureus . 95% of the cultured MSSA were detected by the Detect-Ready MRSA Assay. The assay detected MSSA in nearly twice the number of samples than the culture method. This may either be due to false positive PCRs, or to a lack of selectivity of the here uses culture method for S. aureus. Columbia agar enables the growth of numerous bacteria which can overgrow small amounts of MSSA which are then not detected. A comparison of a selective MSSA agar with the Detect-Ready Assay would be an interesting approach for the future but could not be conducted in scope of this study.
In this study PCR based methods were compared to the direct plating on chromogenic agar. Since not performed under routine conditions neither, no broth-enriched culture was used in the study. Thus, our study is limited by the possibility that broth-enrichement would have lead to more culture positive results. PCR results defined here as false positive could also reflect a detection of non viable MRSA. The reason for choosing the routine culture method as standard was our demand to focus on patients colonised with a considerable amount of still viable MRSA, which determines its transmissibility.
Our study is further limited by the fact, that the false positive samples (amplicons) could not be characterized by molecular methods to examine whether other genetic abnormalities lead to a positive PCR result. No additional data concerning former antibiotic treatment was retrieved for the patients, so it could not be excluded that the detection of false positive samples was due to prior antibiotic treatment, nor to what extent. In contrast to our results, Peterson et al  showed, a higher sensitivity of 95% of the the LightCycler® MRSA Advanced Test in comparison to direct culture, possibly due to an exclusion of patients with antibiotic therapy active against MRSA.
In conclusion, our data show that the LightCycler® MRSA Advanced Test demonstrated a better clinical sensitivity compared to the Detect-Ready® MRSA Kit. We would recommend additional cultural testing in a clinical setting to close the diagnostic gap and to avoid false results. With the CHROMagar MRSA the majority of the positive results (95%) were achieved already after 24 hours, demonstrating this culture based test as a relatively fast, cheep and reliable screening method in situations where no immediate results are needed.,.
Conceived and designed the experiments: VS OS FM. Performed the experiments: AA JL OK KS IW. Analyzed the data: VS OS FM SM. Wrote the paper: JL AA OS.
- 1. Lowy FD (1998) Staphylococcus aureus infections. N Engl J Med 339: 520–532. doi: 10.1056/NEJM199808203390806
- 2. Stewart GT, Holt RJ (1963) Evolution of natural resistance to the newer penicillins. Br Med J 1: 308–311. doi: 10.1136/bmj.1.5326.308
- 3. Kresken M, Hafner D (1999) Drug resistance among clinical isolates of frequently encountered bacterial species in central Europe during 1975–1995. Study Group Bacterial Resistance of the Paul-Ehrlich-Society for Chemotherapy. Infection 27 Suppl 2: S2–S8. doi: 10.1007/BF02561661
- 4. Kresken M, Hafner D, Schmitz FJ, Wichelhaus TA (2003) Resistenzsituation bei klinisch wichtigen Infektionserregern gegenüber Antibiotika in Deutschland ind im mitteleuropäischen Raum. Bericht über Ergebnisse einer multizentrischen Studie der Arbeitsgemeinschaft Empfindlichkeitsprüungen & Resistenz der Paul-Ehrlich-Gesellschaft für Chemotherapie e.V. aus dem Jahre 2001. Antiinfectives Intelligence
- 5. KISS (2010) KISS Krankenhaus-Infektions-Surveillance-System.Available: www.nrz-hygiene.de/fileadmin/nrz/module/mrsa/199701_201104_MRSA_reference.pdf
- 6. Hsueh PR, Teng LJ, Chen WH, Pan HJ, Chen ML, et al. (2004) Increasing prevalence of methicillin-resistant Staphylococcus aureus causing nosocomial infections at a university hospital in Taiwan from 1986 to 2001. Antimicrob Agents Chemother 48: 1361–1364. doi: 10.1128/AAC.48.4.1361-1364.2004
- 7. Klevens RM, Edwards JR, Tenover FC, McDonald LC, Horan T, et al. (2006) Changes in the epidemiology of methicillin-resistant Staphylococcus aureus in intensive care units in US hospitals, 1992–2003. Clin Infect Dis 42: 389–391. doi: 10.1086/499367
- 8. Styers D, Sheehan DJ, Hogan P, Sahm DF (2006) Laboratory-based surveillance of current antimicrobial resistance patterns and trends among Staphylococcus aureus: 2005 status in the United States. Ann Clin Microbiol Antimicrob 5: 2. doi: 10.1186/1476-0711-5-2
- 9. Tiemersma EW, Bronzwaer SL, Lyytikainen O, Degener JE, Schrijnemakers P, et al. (2004) Methicillin-resistant Staphylococcus aureus in Europe, 1999–2002. Emerg Infect Dis 10: 1627–1634. doi: 10.3201/eid1009.040069
- 10. Wertheim HF, Vos MC, Boelens HA, Voss A, Vandenbroucke-Grauls CM, et al. (2004) Low prevalence of methicillin-resistant Staphylococcus aureus (MRSA) at hospital admission in the Netherlands: the value of search and destroy and restrictive antibiotic use. J Hosp Infect 56: 321–325. doi: 10.1016/j.jhin.2004.01.026
- 11. Cosgrove SE, Sakoulas G, Perencevich EN, Schwaber MJ, Karchmer AW, et al. (2003) Comparison of mortality associated with methicillin-resistant and methicillin-susceptible Staphylococcus aureus bacteremia: a meta-analysis. Clin Infect Dis 36: 53–59. doi: 10.1086/345476
- 12. Cosgrove SE, Qi Y, Kaye KS, Harbarth S, Karchmer AW, et al. (2005) The impact of methicillin resistance in Staphylococcus aureus bacteremia on patient outcomes: mortality, length of stay, and hospital charges. Infect Control Hosp Epidemiol 26: 166–174. doi: 10.1086/502522
- 13. Coia JE, Duckworth GJ, Edwards DI, Farrington M, Fry C, et al. (2006) Guidelines for the control and prevention of meticillin-resistant Staphylococcus aureus (MRSA) in healthcare facilities. J Hosp Infect 63 Suppl 1: S1–44. doi: 10.1016/j.jhin.2006.01.001
- 14. Robotham JV, Graves N, Cookson BD, Barnett AG, Wilson JA, et al. (2011) Screening, isolation, and decolonisation strategies in the control of meticillin resistant Staphylococcus aureus in intensive care units: cost effectiveness evaluation. BMJ 343: d5694. doi: 10.1136/bmj.d5694
- 15. Krinko RKI (2008) Kommentar zu den “Empfehlungen zur Prävention und Kontrolle von MRSA-Stämmen in Krankenhäusern und anderen medizinischen Einrichtungen”. Epidemiologisches Bulletin 42: 363–364. doi: 10.1016/j.khinf.2008.10.020
- 16. Hamdan-Partida A, Sainz-Espunes T, Bustos-Martinez J (2010) Characterization and persistence of Staphylococcus aureus strains isolated from the anterior nares and throats of healthy carriers in a Mexican community. J Clin Microbiol 48: 1701–1705. doi: 10.1128/JCM.01929-09
- 17. Lauderdale TL, Wang JT, Lee WS, Huang JH, McDonald LC, et al. (2010) Carriage rates of methicillin-resistant Staphylococcus aureus (MRSA) depend on anatomic location, the number of sites cultured, culture methods, and the distribution of clonotypes. Eur J Clin Microbiol Infect Dis 29: 1553–1559. doi: 10.1007/s10096-010-1042-8
- 18. Ringberg H, Cathrine PA, Walder M, Hugo Johansson PJ (2006) The throat: an important site for MRSA colonization. Scand J Infect Dis 38: 888–893. doi: 10.1080/00365540600740546
- 19. Malhotra-Kumar S, Abrahantes JC, Sabiiti W, Lammens C, Vercauteren G, et al. (2010) Evaluation of chromogenic media for detection of methicillin-resistant Staphylococcus aureus. J Clin Microbiol 48: 1040–1046. doi: 10.1128/JCM.01745-09
- 20. Boye K, Bartels MD, Andersen IS, Moller JA, Westh H (2007) A new multiplex PCR for easy screening of methicillin-resistant Staphylococcus aureus SCCmec types I–V. Clin Microbiol Infect 13: 725–727. doi: 10.1111/j.1469-0691.2007.01720.x
- 21. Hiramatsu K, Katayama Y, Yuzawa H, Ito T (2002) Molecular genetics of methicillin-resistant Staphylococcus aureus. Int J Med Microbiol 292: 67–74. doi: 10.1078/1438-4221-00192
- 22. Boyce JM, Havill NL (2008) Comparison of BD GeneOhm methicillin-resistant Staphylococcus aureus (MRSA) PCR versus the CHROMagar MRSA assay for screening patients for the presence of MRSA strains. J Clin Microbiol 46: 350–351. doi: 10.1128/JCM.02130-07
- 23. Cuny C, Witte W (2005) PCR for the identification of methicillin-resistant Staphylococcus aureus (MRSA) strains using a single primer pair specific for SCCmec elements and the neighbouring chromosome-borne orfX. Clin Microbiol Infect 11: 834–837. doi: 10.1111/j.1469-0691.2005.01236.x
- 24. de San N, Denis O, Gasasira MF, De MR, Nonhoff C, et al. (2007) Controlled evaluation of the IDI-MRSA assay for detection of colonization by methicillin-resistant Staphylococcus aureus in diverse mucocutaneous specimens. J Clin Microbiol 45: 1098–1101. doi: 10.1128/JCM.02208-06
- 25. Huletsky A, Giroux R, Rossbach V, Gagnon M, Vaillancourt M, et al. (2004) New real-time PCR assay for rapid detection of methicillin-resistant Staphylococcus aureus directly from specimens containing a mixture of staphylococci. J Clin Microbiol 42: 1875–1884. doi: 10.1128/JCM.42.5.1875-1884.2004
- 26. Peterson LR, Liesenfeld O, Woods CW, Allen SD, Pombo D, et al. (2010) Multicenter evaluation of the LightCycler methicillin-resistant Staphylococcus aureus (MRSA) advanced test as a rapid method for detection of MRSA in nasal surveillance swabs. J Clin Microbiol 48: 1661–1666. doi: 10.1128/JCM.00003-10
- 27. Fodrie TY, Allen SD, Blue DE, Gary P, Cheng L (2009) Evaluation of a real-time polymerase chain reaction assay using analyte-specific reagents for detection of methicillin-resistant Staphylococcus aureus nasal carriage. Anal Quant Cytol Histol 31: 410–416.
- 28. Garcia-Alvarez L, Holden MT, Lindsay H, Webb CR, Brown DF, et al. (2011) Meticillin-resistant Staphylococcus aureus with a novel mecA homologue in human and bovine populations in the UK and Denmark: a descriptive study. Lancet Infect Dis 11: 595–603. doi: 10.1016/S1473-3099(11)70126-8
- 29. Shore AC, Deasy EC, Slickers P, Brennan G, O'Connell B, et al. (2011) Detection of staphylococcal cassette chromosome mec type XI carrying highly divergent mecA, mecI, mecR1, blaZ, and ccr genes in human clinical isolates of clonal complex 130 methicillin-resistant Staphylococcus aureus. Antimicrob Agents Chemother 55: 3765–3773. doi: 10.1128/AAC.00187-11
- 30. Bartels MD, Boye K, Rohde SM, Larsen AR, Torfs H, et al. (2009) A common variant of staphylococcal cassette chromosome mec type IVa in isolates from Copenhagen, Denmark, is not detected by the BD GeneOhm methicillin-resistant Staphylococcus aureus assay. J Clin Microbiol 47: 1524–1527. doi: 10.1128/JCM.02153-08
- 31. Corkill JE, Anson JJ, Griffiths P, Hart CA (2004) Detection of elements of the staphylococcal cassette chromosome (SCC) in a methicillin-susceptible (mecA gene negative) homologue of a fucidin-resistant MRSA. J Antimicrob Chemother 54: 229–231. doi: 10.1093/jac/dkh284
- 32. Deplano A, Tassios PT, Glupczynski Y, Godfroid E, Struelens MJ (2000) In vivo deletion of the methicillin resistance mec region from the chromosome of Staphylococcus aureus strains. J Antimicrob Chemother 46: 617–620. doi: 10.1093/jac/46.4.617
- 33. Francois P, Bento M, Renzi G, Harbarth S, Pittet D, et al. (2007) Evaluation of three molecular assays for rapid identification of methicillin-resistant Staphylococcus aureus. J Clin Microbiol 45: 2011–2013. doi: 10.1128/JCM.00232-07
- 34. Bode LG, Kluytmans JA, Wertheim HF, Bogaers D, Vandenbroucke-Grauls CM, et al. (2010) Preventing surgical-site infections in nasal carriers of Staphylococcus aureus. N Engl J Med 362: 9–17. doi: 10.1056/NEJMoa0808939
- 35. Luteijn JM, Hubben GA, Pechlivanoglou P, Bonten MJ, Postma MJ (2011) Diagnostic accuracy of culture-based and PCR-based detection tests for methicillin-resistant Staphylococcus aureus: a meta-analysis. Clin Microbiol Infect 17(2):146–54 doi: 10.1111/j.1469-0691.2010.03202.x. doi: 10.1111/j.1469-0691.2010.03202.x
- 36. Arcenas RC, Spadoni S, Mohammad A, Kiechle FL, Walker K, et al. (2012) Multicenter Evaluation of the LightCycler MRSA Advanced Test, the Xpert MRSA Assay, and MRSASelect Directly Plated Culture with Simulated Workflow Comparison for the Detection of Methicillin-Resistant Staphylococcus aureus in Nasal Swabs. J Mol Diagn 14(4):367–75. doi: 10.1016/j.jmoldx.2012.01.015